|Product Name:||ORP Electrode Sensor||Model:||ASR301-VYD05|
|Working Temperature:||0.0~60.0℃||Reference Electrode:||Ag/AgCl|
|Detection Heads:||Ring-shaped Platinum||Application:||Industrial Water|
|Cleaning:||Cleaning Interface||Temp. Unit:||Without|
Ring Shaped Platinum ORP Sensors,
ABS Shaft ORP Sensor,
Water Monitoring PH ORP Sensors
±2000mV 0~60℃ ABS ORP Sensor For Swimming Pool Water Monitoring
Almost all ORP depends on chloride ion (Cl-) and pH (H+), just same as it depends on hypochlorous acid (chlorine in water).
Any changes in chlorine concentration or pH value can affect ORP . Therefore, in order to measure chlorine accurately, chloride ion and pH value must be measured with high accuracy, or carefully controlled to constant values.
ORP is not compensated for any change in temperature, and the vast majority of responses are pH-dependent.
Please click the catalog of ORP controller here:
◆ ORP composite electrode, the body adopts reinforced ABS plastic shell encapsulation
◆ The electrode is equipped with a protective cover with its own cleaning interface
◆ The electrode is divided into two basic types: ring platinum detection head and flat platinum detection head;
◆ Ring-shaped platinum has a wide response range and is suitable for sewage environment;
◆ Platinum is easier to clean and polish, suitable for water purification environment;
◆ The reference adopts a large-area annular porous Teflon diaphragm
◆ The reference system is designed with double salt bridge, the reference electrode is Ag/AgCl, filled with KCL solid gel.
Water treatment monitoring
Drinking water,surface water,water quality analyze,ground water,ect.
Industrial process monitoring and dosing control
Chemical,petroleum,paper,food and medicine,PCB manufacturing,etc.
1. Does ORP measurement require temperature compensation?
A: ORP measurement is not temperature controlled, so the ORP sensor does not require temperature compensation.
2. How do we choose the ORP electrode when the oxidation level is high?
A: In occasions where the oxidation level is high, that is, the ORP value is high, platinum electrodes are recommended, otherwise gold electrodes are recommended. ORP measurements usually use the above two inert materials as indicator electrodes.
3. What is the wire length of the ORP electrode?
A: The lead wire of this electrode is 5m by default, other lengths need to be customized, contact our customer service, the price will vary with the change of length.
4. What should ORP electrodes be used with?
A: ORP electrode cannot be used alone, it must be connected with PH/ORP controller. We have a matching controller with a suitable price, and you can choose.
5: Are you trading company or manufacturer ?
online, portable(single-parameter and multi-parameter), water analysis instruments
9: Do you offer the OEM service?
A: Yes, we can offer OEM service and design the logo marking for you. But the minimum quantity is 50pcs for the first order .
Contact Person: Yuki Fu