|Product Name:||PH Sensor||Model:||ASP 400-NYD05|
|Working Temp.:||0.0~100.0℃||Working Pressure::||0~6Bar|
|Reference System:||Double Salt Bridge||Reference Reference Electrode:||Ag/AgCl|
ABS Material PH Probe,
Analog Type PH Probe,
Pure Water PH ORP Sensors
Pipeline installation 60℃ ABS analog type pH probe for Pure water
The reference electrode portion of the pH electrode can be connected to the sample in a variety of ways. These formats have been refined over the years because different samples are measured and the requirements for the electrodes are also different.
The standard junction is, and is the simplest, the ceramic junction. It is made of a porous ceramic core passed through the electrode glass rod. This porous ceramic core material ensures that the electrolyte seeps out slowly and prevents rapid draining. This type of junction is ideal for measuring aqueous solutions;
Please click the user manual of pH sensor here:
◆ pH composite electrode, the body adopts high-performance ABS plastic shell encapsulation
◆ Sturdy structure, with installation threads at the beginning and end, easy to install;
◆ The electrode is equipped with a protective cover with its own cleaning interface;
◆ The indicator electrode adopts a low-impedance hemispherical glass head, which has high strength and is not easily broken;
◆ The reference adopts a large-area annular porous Teflon membrane, which has fast penetration speed, fast response, not easy to block, and easy to clean;
◆ The reference system is designed with double salt bridge, the reference electrode is Ag/AgCl, and the reference is filled with saturated KCL+excess KCL crystal;
Water treatment monitoring
Drinking water,surface water,water quality analyze,ground water,ect.
Industrial process monitoring and dosing control
Chemical,petroleum,paper,food and medicine,PCB manufacturing,etc.
The pH value potentiometric method uses the potential difference of glass electrodes for pH measurement.
The pH electrode is usually composed of an indicator electrode and a reference electrode.
If the internal hydrogen ion concentration of the pH glass bulb is lower than the outside, the measured solution is recorded as acidic and the pH value is lower than 7; if the two are the same, the solution is neutral and the pH value is equal to 7; if the internal hydrogen ion concentration is higher than externally, the measured solution is alkaline and the pH value is higher than 7.
4. What is the wire length of the pH electrode?
A: The lead wire of this electrode is 5m by default, other lengths need to be customized, contact our customer service, the price will vary with the change of length.
5. How often must the pH sensor be calibrated?
A: Depends on the type of sample:
Dirty samples and non-aqueous samples require more frequent calibration.
Depends on the required accuracy:
For very accurate measurements, at least daily calibration is required.
Depends on the quality of the electrode:
Old electrodes require more frequent calibration.
Contact Person: Yuki Fu