|Product Name:||PH Sensor||Model:||ASP 400-VYD05|
|PH:||0~14pH||Materials In Contact With Liquid:||PPS+Glass+Teflong|
|Working Pressure::||0~6Bar||Working Temperature:||0.0~100.0℃|
|Reference Reference Electrode:||Ag/AgCl||Reference System:||Double Salt Bridge|
|Temp. Unit:||Vacant||Cleaning:||Cleaning Interface(Optional)|
High Temp Resistant PH Electrode,
PPS Shaft PH ORP Sensors,
Sewage Water PH Electrode
0~100℃ pH electrode with High temp. resistant for Sewage water
The pH electrode determines the accuracy of pH measurement, so pH electrode status is critical. Therefore, in order to prolong the life of the pH electrode and the accuracy of the results, the daily maintenance of the electrode is very important. If the electrode is not cleaned after use or electrode maintenance is neglected for a long time, the electrode will no longer be accurate and the accuracy of the system will quickly degrade. This is manifested by a gradual decrease in the slope of the electrode. When the slope drops to 50mV/pH (equivalent to 85% of the slope) or the zero point exceeds ± 30 mV, some treatment methods can restore the electrode to the expected performance, but for the accuracy of the measurement results, it is necessary to replace a new electrode.
Please click the user manual of pH sensor here:
◆ pH composite electrode, the body adopts high-performance ABS plastic shell encapsulation
◆ Sturdy structure, with installation threads at the beginning and end, easy to install;
◆ The electrode is equipped with a protective cover with its own cleaning interface;
◆ The indicator electrode adopts a low-impedance hemispherical glass head, which has high strength and is not easily broken;
◆ The reference adopts a large-area annular porous Teflon membrane, which has fast penetration speed, fast response, not easy to block, and easy to clean;
◆ The reference system is designed with double salt bridge, the reference electrode is Ag/AgCl, and the reference is filled with saturated KCL+excess KCL crystal;
Water treatment monitoring
Drinking water,surface water,water quality analyze,ground water,ect.
Industrial process monitoring and dosing control
Chemical,petroleum,paper,food and medicine,PCB manufacturing,etc.
The pH value potentiometric method uses the potential difference of glass electrodes for pH measurement.
The pH electrode is usually composed of an indicator electrode and a reference electrode.
If the internal hydrogen ion concentration of the pH glass bulb is lower than the outside, the measured solution is recorded as acidic and the pH value is lower than 7; if the two are the same, the solution is neutral and the pH value is equal to 7; if the internal hydrogen ion concentration is higher than externally, the measured solution is alkaline and the pH value is higher than
The lead wire of this electrode is 5m by default, other lengths need to be customized, contact our customer service, the price will vary with the change of length.
5. How often must the pH sensor be calibrated?
A: Depends on the type of sample:
Dirty samples and non-aqueous samples require more frequent calibration.
Depends on the required accuracy:
For very accurate measurements, at least daily calibration is required.
Depends on the quality of the electrode:
Old electrodes require more frequent calibration.
6: What's your business range?
A: We manufacture Water quality analysis. The monitoring parameters of our water quality analyzer main include: PH, ORP, dissolved oxygen, conductivity/TDS/salinity, turbidity, SS, Chlorophyll, Blue-green algae, COD, NH4-N and so on.
7: May I visit your factory?
A: Of course, our factory is located in suzhou city , welcome your arrival.
Contact Person: Yuki Fu