|Product Name:||PH Sensor||Model:||ASP 400-VYD05|
|PH:||0~14pH||Materials In Contact With Liquid:||PPS+Glass+Teflong|
|Working Pressure::||0~6Bar||Working Temperature:||0.0~100.0℃|
|Reference Reference Electrode:||Ag/AgCl||Reference System:||Double Salt Bridge|
|Temp. Unit:||Vacant||Cleaning:||Cleaning Interface(Optional)|
Anti Pollution PH Probe Probe,
Auto Cleaning Interface PH Probe Sensor,
Flow Through PH Electrode Electrode
Online Pps Material PH Probe With High Temperature Resistant For Waste Water
Signal output: RS-485 (Modbus protocol). Used for environmental water quality monitoring, acid/alkali/salt solution, chemical reaction process and industrial production process, it can meet the requirements of online pH measurement for most industrial applications.
Please click the pH sensor user manual here:
ASP400T pH sensor technical data.pdf
◆ pH composite electrode, the body adopts high-performance ABS plastic shell encapsulation
◆ Sturdy structure, with installation threads at the beginning and end, easy to install;
◆ The electrode is equipped with a protective cover with its own cleaning interface;
◆ The indicator electrode adopts a low-impedance hemispherical glass head, which has high strength and is not easily broken;
◆ The reference adopts a large-area annular porous Teflon membrane, which has fast penetration speed, fast response, not easy to block, and easy to clean;
◆ The reference system is designed with double salt bridge, the reference electrode is Ag/AgCl, and the reference is filled with saturated KCL+excess KCL crystal;
Water treatment monitoring
Drinking water,surface water,water quality analyze,ground water,ect.
Industrial process monitoring and dosing control
Chemical,petroleum,paper,food and medicine,PCB manufacturing,etc.
1. What is themeasurement principle of this pH electrode?
A: The pH value potentiometric method uses the potential difference of glass electrodes for pH measurement.
2. How is this glass pH electrode composed?
A: The pH electrode is usually composed of an indicator electrode and a reference electrode.
3. How to judge the pH of the solution?
A: If the internal hydrogen ion concentration of the pH glass bulb is lower than the outside, the measured solution is recorded as acidic and the pH value is lower than 7; if the two are the same, the solution is neutral and the pH value is equal to 7; if the internal hydrogen ion concentration is higher than externally, the measured solution is alkaline and the pH value is higher than
4. What is the wire length of the pH electrode?
A: The lead wire of this electrode is 5m by default, other lengths need to be customized, contact our customer service, the price will vary with the change of length.
5. How often must the pH sensor be calibrated?
A: Depends on the type of sample:
Dirty samples and non-aqueous samples require more frequent calibration.
Depends on the required accuracy:
For very accurate measurements, at least daily calibration is required.
Depends on the quality of the electrode:
Old electrodes require more frequent calibration.
6: What's your business range?
A: We manufacture Water quality analysis. The monitoring parameters of our water quality analyzer main include: PH, ORP, dissolved oxygen, conductivity/TDS/salinity, turbidity, SS, Chlorophyll, Blue-green algae, COD, NH4-N and so on.
7: May I visit your factory?
A: Of course, our factory is located in suzhou city , welcome your arrival.
Contact Person: Yuki Fu